Specifications Relating to Non-automatic Weighing Devices (1998)
52. All markings shall be located
(a) in a clearly visible place on a part of the weighing device or on a descriptive plate fixed to it; and
(b) near the weight display, in the case of the marking for
(i) the maximum capacity,
(ii) the actual scale interval, and
(iii) if different from the actual scale interval, the verification scale interval.
- Date modified: