Weights and Measures Regulations
302 A capacity indicator shall only be installed in a tank at any level where a quantity of liquid equal to the limit of error referred to in section 325 for the indicated quantity causes the liquid at that level to rise by at least 2.5 mm or 3/32 of an inch.
- SOR/89-570, s. 6(F)
- SOR/90-118, s. 41
- Date modified: