Specifications Relating to Non-automatic Weighing Devices (1998)
49 (1) The information set out in column I of the table to this section shall be marked, using appropriate words or the symbols set out in column II of that table, on the weighing devices and modules described in columns III to VI of that table.
Column I Column II Column III Column IV Column V Column VI Information Symbols Modules in the same housing Indicating elements tested separately Weighing and load-receiving elements tested separately Other equipment and accessories approved separately Manufacturer, applicant or importer’s name or trade mark – – – yes yes yes yes Model number – – – yes yes yes yes Distinctive serial number – – – yes yes yes yes Approval number – – – yes yes yes yes Accuracy class , , , or yes yes yes – – – Maximum capacity Max yes yes yes – – – Verification scale interval e yes yes – – – – – – Actual scale interval, if different from the verification scale interval d yes yes – – – – – – Minimum verification scale interval emin – – – – – – yes – – – Maximum number of verification scale interval nmax – – – yes yes – – – Temperature range, if different from -10°C to 40°C – – – yes yes yes – – –
(2) In this section,
(a) Modules in the same housing includes
(b) Indicating elements tested separately includes individually-approved indicating elements that may be attached to other approved modules;
(c) Weighing and load-receiving elements tested separately includes individually-approved weighing and load-receiving elements that may be attached to approved modules; and
(d) Other equipment and accessories approved separately includes other equipment and accessories that perform metrological functions and that can be tested and approved separately.
(3) In the case of the elements described in column IV of the table to this section, the maximum capacity and the actual scale interval shall be marked as required by paragraph 52(b) at the time they are attached to other modules to form a weighing device.
- SI/2008-81, ss. 9, 19
- Date modified: