Specifications Relating to Non-automatic Weighing Devices (1998)
45 A weighing device that is operating in static mode and that has a digital indicating element shall permit zeroing, taring, printing or the storage of weight values only when the weight indication is stable within
(a) ± 3 verification scale intervals, in the case of a device with a maximum capacity of more than 2 000 kg (5 000 pounds); and
(b) ± 1 verification scale interval, in the case of all other weighing devices.
- SI/2008-81, s. 7
- Date modified: