PART VSpecifications Relating to Devices (continued)
DIVISION XIVolumetric Liquid Meters (continued)
Installation and Use
271 A meter shall be properly suited to the particular installation in which it is to be used, with respect to pressure, rate of flow, temperature, compatibility with the characteristics of the liquid to be measured and method of use so that, for all normally occurring variations in operating conditions, measurement by the meter remains within the applicable limits of error.
- SOR/89-570, s. 6(F)
272 The equipment and accessories installed with a meter shall be selected and installed so as to minimize their interference with the proper operation and accuracy of measurement of the meter.
273 The pump suction piping installed with a meter shall be as short and unencumbered as possible and all liquid supply systems associated with a meter shall be arranged to minimize the ingress of air, water or other foreign matter, including any liquid other than the one the meter is intended to measure.
274 A meter and all equipment and accessories associated with it in an installation shall be installed so as to minimize the evolution of dissolved air and gases and to minimize the vaporization of the liquid prior to or during measurement or in the delivery conduit.
275 (1) A meter shall be installed so that it can only be operated when the liquid in the meter is at or above atmospheric pressure, and in any installation where the liquid pressure could fall below atmospheric pressure, a suitable vacuum breaker shall be installed at the outlet of the meter.
(2) A meter shall be operated only when the liquid in the meter is at or above atmospheric pressure.
276 When a meter is installed in a measuring system, the system shall be equipped with effective automatic means to minimize the passage of air or vapour through the meter.
277 A meter shall have installed immediately upstream therefrom a screen, strainer, filter, integral filter or such other approved means as will prevent foreign matter from passing through the meter.
278 For the purposes of sections 276 and 277, a close-coupled strainer and automatic air release valve is sufficient in meter installations where only free air or vapour is to be removed from the liquid.
279 The air or vapour eliminator installed with a meter shall minimize the passage of air or vapour through the measuring element of the meter when the supply of liquid is exhausted or appropriate means shall be provided to automatically stop the flow of liquid or to indicate to the person using the meter that air or vapour is passing through the meter.
280 Sealed flow control valves or other means shall be installed with a meter to limit the flow through the meter to its rated maximum flow.
281 Unless a meter is equipped with a reversing register and totalizer in accordance with section 250, it shall be installed so as to minimize any reverse flow through it or the piping or equipment upstream or downstream from it that could affect the accuracy of its measurement.
(2) A meter shall be installed so that the piping between the meter and the transfer point remains full of liquid during and after a transaction unless the liquid, because of its nature, cannot remain in the meter, and
(a) in the case of a meter installed to deliver a quantity of liquid, so that the piping downstream of the transfer point shall empty at the completion of the transaction; and
(b) in the case of a meter installed to receive a quantity of liquid, so that the piping upstream of the transfer point shall empty at the completion of the transaction.
(3) Subject to section 289 and except as otherwise allowed by any additional specifications established pursuant to section 27, a meter shall be installed
(a) so that all piping downstream of the meter, where the meter is installed to deliver a quantity of liquid, or all piping upstream of the meter, where the meter is installed to receive a quantity of liquid, may be readily examined; and
(b) with only one delivery outlet unless the meter is installed
(i) so that diversion of flow is readily apparent to the purchaser or the purchaser’s agent or mandatary by automatic means such as visible valves or lights with explanatory signs that indicate which outlets are in operation,
(ii) to fuel or defuel aircraft, or
(iii) to be used exclusively to load vehicle-mounted propane tanks with a capacity greater than 5 000 L.
- SOR/90-118, s. 38
- SOR/2014-111, ss. 35(E), 46
283 A meter, other than a slow-flow meter, slow-flow dispenser, dispenser of the service station type for motor fuel or a meter equipped with a preset valve, shall have a quick-acting valve installed near its outlet for examination purposes.
- SOR/93-234, s. 2(F)
- SOR/2014-111, s. 46
284 A meter, other than a slow-flow meter, shall be installed so that it has a convenient means for delivering the measured product when the meter installation is examined.
- SOR/2014-111, s. 36
285 Any automatic valve that can have a throttling effect on the flow of liquid shall be installed downstream from the meter.
286 (1) Unless otherwise authorized by specifications established by the Minister, the outlet end of the delivery hose of a pump-supplied meter shall be equipped with a spring-loaded anti-drain valve or its equivalent.
(2) Subsection (1) does not apply to a meter with collapsible hoses used on aircraft refuellers or to a meter designed for defuelling aircraft.
287 The discharge hose installed with a meter supplied by gravity head may be of the dry hose type without a shut-off valve at the discharge end, but it shall be of such length, size and stiffness as to ensure complete drainage and there shall be means provided in the meter system to show the empty condition of the hose and to prevent the meter from draining.
288 Where it is necessary
(a) to limit changes in the viscosity of, or the solubility of material dissolved in, the liquid to be measured by the meter, or
(b) to limit the rate of change of the temperature or pressure of the liquid to be measured by the meter when the meter is equipped with an automatic compensator,
appropriate automatic controls shall be installed in the meter system to permit the meter to measure accurately.
- SOR/90-118, s. 39
(a) are used
(b) have adequately interlocked manual or automatic valves with means for leakage detection incorporated in the meter systems to prevent the diversion of measured liquid.
(2) The piping on the downstream side of the meters may be concealed where the meters have more than one outlet and are used exclusively for refuelling trucks.
- SOR/2005-297, s. 21
290 A meter shall be operated only at flow rates between the rated minimum and maximum flow rates marked on the name plate of the meter.
291 If any part of a meter system is drained for any reason the entire system shall be re-primed and re-filled and the register set to zero before the meter is used for the next measurement.
292 A flow transmitter for input to an electronic data processing system may be attached to a liquid meter and the output from the data processing system used for invoicing purposes if the purchaser of the liquid being measured or their agent or mandatary is given a printed copy of the data recorded by the data processing system before they leave the trader’s premises.
- SOR/2014-111, s. 37
293 A flow transmitter may be installed at a key-operated self-service meter installation to drive the auxiliary accumulative totalizers for individual customers if the purchaser of the liquid being measured is able to visually check the advance of their totalizer against the indication of the primary indicator at the time of delivery of the liquid.
- SOR/2014-111, s. 37
294 Where a meter mounted on a vehicle tank is used in circumstances where the purchaser does not normally view the meter during delivery of a liquid, the meter shall be equipped with an interlocked ticket printer that shall be used for each delivery of the liquid.
295 Tickets used with a meter equipped with an automatic temperature or other compensator shall bear statements that the volume indicated thereon has been adjusted to correspond to an equivalent volume at a standard condition stated on the ticket.
- Date modified: